The BR-MV14 ventilator is gas driven by medical compressed oxygen and compressed air. In the inspiratory phase, two streams of compressed gas (compressed oxygen and compressed air) flow into a high performance air-oxygen mixer to form a mix of oxygen and air with a certain pressure.
10.4'' LCD Display
pneumatically powered and electrically controlled system
Closed; Semi-Closed; Semi-Open
Fitted with 4 nos anti-static rubber castors; two of which are lockable for braking and easy maneuverability with foot operated brake provisions
Medical oxygen and nitrous oxide with a pressure ranging from 280~600kpa
gas concentration output
21% ~ 100% adjustable
Maximum Minute ventilation
≥18L / min, permissible deviation: ±20 %
A/C, SIPPV、IPPV、SIMV、PSIMV、CPAP、PCV、PEEP、SIGN、SPONT, MANUAL
: ≤30 Ml/kPa
1 ~ 12 times/min, permissible deviation: ±15%.
Tidal Volume range
0~2000ml resolution ratio: 10ml
Maximum Safety Pressure
one deep breath per 80 controlled respirations, the inspiration time is 1.5 times of the setting point and the permissible error is -10% or 15%.
Pressure limit range:
Airway pressure alarm: Audible and visual and with yellow and red color indicating
upper alarm: 0.9 kPa ~ 5.4 kPa
Ventilation Volume alarm:
≥ 18L/ MIN
Power Supply Alarm
Ac/dc power supply are after failing to send out alarm immediately
Alarm Time: keep >120s
The airway pressure continues to be higher than 15 hPa ±1 hPa for 15s±1s, then the machine will raise an audible alarm, the pressure will be displayed in red and the continuous high pressure red alarm wording is displayed on the screen of the anesthetic respirator.
10 ~ 40oC
no higher than 80 %
860 hPa ~ 1060 hPa
100-120 Vac, 50/60 Hz;
Air source requirement:
Medical oxygen and laughing gas with a rated pressure ranging from 0.3 to 0.5MPa.
Attention: the AC power supply used for the anesthesia machine must be well grounded.
Attention: the anesthesia machine used must be equipped with a carbon dioxide monitor complying with ISO 9918:1993, an oxygen monitor complying with ISO 7767:1997 and an expiratory gas volume monitor complying with 18.104.22.168 of Medical Electrical Equipment Part II: Special Requirements for Safety and Basic Performance of Anesthesia System.
-15oC ~ +50oC
no higher than 95 %
86 kPa ~ 106 kPa.
It should be stored in a room without corrosive gas and well ventilated
comply with the requirement of GB/T 15464
Between the packaging box and product, soft material with appropriate thickness provided to prevent loosening and mutual friction during transportation
Moisture protection and rain protection to ensure that product is protected from natural damage.
Safty & Alarm
Gas source pressure
Immediate alarm; “high level alarm” indicator flickers raising an audible alarm.
Airway Pressure Alarm
The airway pressure exceeds 15hpa. The maximum delay is no longer than 17s. The “high level alarm” indicator flickers raising an audible alarm.
Tidal Volume Alarm
Lower than 50ml, The alarm will be raised after one breathing cycle.
Ventilation Volume Alarm
Lower: 0~12L/Min; Up: 18L/ Min
Battery Voltage Alarm
It alrms when the battery voltage is lower than 10.5V.
Power On Self Test
Self test before working, assure it's working normally.